Raster orbital scanning (ROS) is a technique that dramatically increases the signal-to-noise ratio of Raman measurements made
from target materials dispersed on surface enhanced Raman scattering (SERS) substrates. For SERS substrates, rastering a laser
beam across the sample decreases the background and samples many more Raman-active compounds compared to traditional static-beam
techniques. The lower average power created by rastering the beam with ROS dilutes the laser power over a larger sample area.
This lower average power reduces the risk of damaging the substrate, which often increases the background of the Raman spectrum.
A key challenge in acquiring high quality Raman data is reducing the background component of the spectrum, which can obscure
peaks and make sample identification difficult. When analyzing SERS samples, be aware that the target material likely will
be dispersed non-uniformly on the substrate. As a result, the static, tightly-focused laser spot used in traditional Raman
sampling will often poorly overlap the Raman-active material and measure mostly background substrate. Also, a static laser
beam can cause localized heating and damage to the SERS substrate, further increasing the background of the spectrum. ROS
sampling found in Ocean Optics IDRaman reader and IDRaman mini products overcomes these challenges and improves SERS measurements.
ROS Sampling of SERS Substrates
In Figure 1, Raman shift versus wavenumber is shown for a SERS-active nanoparticle membrane tagged with 2-mercaptopyridine-N-oxide
measured at 4 mW of laser power. The benefit of ROS sampling is demonstrated when the spectrum measured with traditional Raman
sampling (emulated by turning ROS sampling off) is compared to the spectrum measured with ROS sampling. When raster scanning
is enabled, the background (represented by the hashed line in the ROS off example) is much lower and peaks are more clearly
resolved. This allows simpler and more confident identification of the target material.
Figure 1: A SERS sample measured at 4 mW of laser power with and without ROS sampling. ROS reduces the background component
of the spectrum by collecting more data from the Raman-active hotspots relative to the background. The elevated background
of a traditional, static laser beam sampling method is illustrated by the hashed line.
Furthermore, the effect of localized heating damage on SERS substrates is demonstrated in Figure 1 via the hashed, horizontal
line on the y-axis. When ROS is not used, the background increases substantially. The constant movement of the laser spot with ROS turned
on (lower spectrum) minimizes sample damage caused by localized heating from a static, focused laser spot. Therefore, ROS
enables the use of higher laser power without degrading the quality of the Raman measurement.
As shown in this application note, the measurement of SERS samples can be greatly improved with the use of ROS. ROS reduces
the background component of SERS spectra and helps to reveal features previously hidden in the high background of SERS substrate
measurements. In addition, ROS samples a larger area and facilitates the use of less expensive, less homogeneous SERS substrates
with the same enhancement of Raman activity. Lastly, ROS enables the use of higher laser power without the risk of damaging
the SERS substrates, increasing the signal enhancement realized by the SERS technique.
830 Douglas Avenue, Dunedin, FL 34698
tel. +1(727) 733-2447, fax: +1(727) 733-3962