The permitted sulfur limit in fuel oils has been decreased to 10 ppm in many regions. This application note demonstrates that the performance of the Rigaku Micro-Z ULS meets the requirement of ASTM D2622-10, in the analysis of ultra-low sulfur in automotive fuels.
Recent developments in ultra-low sulfur (ULS) fuel have improved fuel efficiency and created cleaner exhaust gas. Globally, the permitted sulfur limit in fuel oils has been decreased to 10 ppm in many countries and regions.
For compliance verification, X-ray fluorescence (XRF) spectrometry is the definitive analysis tool for use at distribution terminals and refineries, as well as mobile or stationary testing laboratories. In recent years, there has been an increasing need for an instrument which does not require the use of helium gas, for instances when acquisition or delivery of helium to the analysis site is difficult.The Micro-Z ULS (ultra-low sulfur) is a newly developed sulfur analyzer which does not require helium gas in operation. This application note demonstrates that Micro-Z ULS can meet the requirements of ASTM D2622-10.
The Micro-Z ULS is a benchtop wavelength dispersive X-ray fluorescence (WDXRF) spectrometer with fixed optics optimized for sulfur analysis. The atmosphere in the optics path of the Micro-Z ULS is a vacuum, so that helium gas is not required. The spectrometer is designed to minimize the restrictions in installation such as the need for cooling water, a special power supply, and installation space. The Micro-Z ULS is equipped with an air-cooled 40 W Cr-target X-ray tube and a doubly curved RX-9 analyzing crystal optimized for low concentration sulfur.
The Micro-Z ULS has a built-in operation panel, which enables simple and easy operation for daily analysis. The counting time was 300 s for peak and background measurements in this experiment.
Approximately, 4 mL of each sample was poured into a liquid cell with sample film of 2.5 µm Mylar® film (Chemplex® 100).
Standard and Calibration
The difference between successive test results obtained by the same operator with the same apparatus under constant operation conditions on identical test material would exceed the values calculated by equation 1 only in one case in every 20.
The test results, where the maximum value for each fuel is smaller than r, prove that the performance of the Micro-Z ULS meets the requirement of ASTM D2622-10 for diesel fuel and gasoline.
Low concentration sulfur in petroleum-based fuel can be routinely analyzed with high accuracy and precision on the Micro-Z ULS, a benchtop WDXRF spectrometer with fixed optics, which does not require He gas, external water for cooling the X-ray tube, or special power supply. This application note demonstrates that the performance of the Micro-Z ULS meets the requirement of ASTM D2622-10, which has become strict in the recent versions of ASTM D2622.
(1) ASTM D2622-10(2010), "Standard Test Method for Sulfur in Petroleum Products by Wavelength Dispersive X-ray Fluorescence Spectrometry," ASTM International, p. 12.